Promotional Tickets from our Partners!
Featured: Women in STEM Conference and Award, San Francisco, Sept 10-12, 2017
Featured: PrecisionAg Vision Conference, Phoenix, Oct 10-12, 2017
Featured: SPTechCon, SharePoint Technology Conference, Washington, DC, Nov 12-15, 2017
Boston Professional Events List
3 Hrs Virtual Seminar Design Control Virtual Bootcamp: Hazard and Usability Analysis
Link to Website
The U.S. Food and Drug Administration (FDA) is responsible for regulating the development of and manufacturing processes for medical devices. The FDA also determines the necessary requirements before these medical devices can hit the markets. It’s crucial that FDA design controls are strictly followed in the process of product development.
However, designing a medical device and testing it to prove that it works is not sufficient in the eyes of the FDA to provide a safe product for users. Medical device designers, developers, engineers and manufacturers must be intimately familiar with the FDA’s requirements that ensure a product is safe for use before it reaches the markets.
In this Virtual Boot Camp, Edwin Waldbusser will guide you on how to comply with the FDA’s requirements for designing and testing a medical device.
What is design control?
When does it start?
Why hazard analysis following ISO14971 is the best risk control method
Human factors following new FDA guidance and ISO 62366
Why software validation is more than testing
Session 1: Design Control
Length: 60 minutes
The FDA has determined, through analysis of product recall data, that the majority of recalls were due to a faulty design process, not faulty manufacturing. These recalled products were tested before release and later failed in unanticipated ways that were not considered in the design and testing process. According to the FDA, the factors resulting in the design process that will reduce the chances of an unsafe product include: a well-controlled design process with risk analysis, change control, design reviews, human factors analysis, hardware/software validation and feedback of the risk analysis results into the design process.
This session will answer the important and confusing question of when design control begins in a development process. The session will explain the difference between pre-release and post-release change control. Edwin will further explain the design history file and will discuss a contents checklist. He will also explain the interrelationship between ongoing risk analysis and the design process.
- Reasons for design control
- When design control begins
- Elements of a design control program
- How risk managements fits into design control
- How human factors fits into design control
- Change control
- Understanding validation consists of more than testing
- Design history file
Session-2: Hazard Analysis
Length: 60 minutes
The FDA expects that risk management will be conducted as a part of a product development program. The agency recommends using ISO 14971 as a guide and has accepted it as a recognized standard. Hazard analysis is the most powerful of the risk management tools described in ISO 14971; however, it can be very confusing.
Many new concepts have been introduced. This session will explain these new concepts and will provide you with examples so that the process is clear. One of the techniques described in ISO 14971 is hazard analysis, which is the most powerful of the risk management techniques because it considers risks in normal operation as well as fault conditions. FMEA and FTA consider only fault conditions and are more suited as reliability tools than as product safety tools.
In this session, Edwin will explain in detail the process of conducting a hazard analysis. It will explain confusing terms like “hazard,” hazardous situation,” “harm,” “causative event,” “ALARP,” “risk index” and “residual risk.”
- Explanation of confusing terms
- Step-by-step explanation of hazard analysis process using a template
- Integration of human factors studies into the hazard analysis
- How to deal with residual risk?
- How to integrate human factors studies into the hazard analysis
- How to integrate hazard analysis into the design program
Length: 60 minutes
The FDA will only approve devices that are designed so that it is practically impossible for people to accidentally harm themselves – even if they use the device improperly. The FDA has replaced the term “user error” with “use error.” The FDA considers use error to be a device non-conformity because human factors must be considered in the design process. The burden is on the device designer to create an “idiot-proof” product.
Human factors/usability is the analysis of how people interact with medical devices. This session will explain the process of conducting rigorous human factors studies throughout the design process, integrating it with the device risk analysis and design process, and validating the effectiveness of the studies. Edwin will also discuss various types and methods of human factors analysis. This process conforms to the new ISO 62366 standard and the new FDA Guidance document.
- User error vs. use error
- Human factors studies for design process
- Device risk analysis
- Validation of the effectiveness of human factors studies
Who Will Benefit
Medical Device Engineering Personnel
Medical device designers
QA / RA
Mid- and Senior-level Management
CEOs and CFOs
Sales and Marketing
Other Medical Products Manufacturing Stakeholders
Edwin Waldbusser is a consultant who is retired from industry after 20 years in management of development of medical devices (including five patents).
(Contact us : 844-267-7299 | 954-947-5671
Don't wait. Register TODAY or Email US [email protected]
VIP Life Time Subscription to our Newsletters!